Reduction of the stronger ring artifacts, the ring artifact that is not the center of the image…
Ultra-fast reconstruction function with multi-FPU is available now.
Multi-detectior row CT scan functions are available now.
New CT reconstruction software for helical CT is released
New TomoShop®: High quality CT reconstruction software is released!
Please bookmark this site. We put valuable information about the solution for your CT problem, so why not bookmark it! Thank you.
If you have problems with the reconstruction result from CT, if you are developing or modifying your CT system, and looking for adequate high quality CT reconstruction software, we can provide the best solution for you. Please feel free to contact us.
The problems with X-ray CT reconstruction
There are various problems with achieving the adequate image results using CT system to scan for NDI. The problems are artifacts problems, focusing problems, and others as well as the time duration of CT reconstruction time. Our high quality CT reconstruction software: TomoShop® has been developed for solving those problems with using CT in the various fields.
To begin with, let us show you those problems. Please check the page below.
The right image shows the example of ring artifact. Ring artifact is a common artifact in CT result. Particularly, when the low power X-ray is used to CT scan, its shown on the result significantly.
The left image is the example of metal artifact. Metal artifact is very common artifact when the metals are contained in the samples to be taken CT scan.
Read more ☞ “Problem cases with X-ray CT scan” page
☞ Go to TomoShop® Video Page